The US DHS Cybersecurity & Infrastructure Security Agency (CISA) issued a security advisory to warn of three recently patched flaws in Medtronic Valleylab products that could be exploited to install a
The flaws affect Medtronic Valleylab FT10 and FX8 devices, experts warn that that network connectivity for these systems is often enabled exposing them to remote hack.
“Successful exploitation of these vulnerabilities may allow an attacker to overwrite files or remotely execute code, resulting in a remote, non-root shell on the affected products. By default, the network connections on these devices are disabled.” reads the advisory. “Additionally, the Ethernet port is disabled upon reboot. However, it is known that network connectivity is often enabled.”
The first vulnerability, tracked as CVE-2019-13543, is related to the use of
The CVE-2019-13543 vulnerability has received a base score of 5.8.
The second flaw discovered in the Medtronic Valleylab products (CVE-2019-13539) ties the use of the
Another vulnerability is related to the use of a vulnerable version of the
Affected products are Valleylab Exchange Client version 3.4 and below, Valleylab FT10 Energy Platform (VLFT10GEN) software version 4.0.0 and below, and Valleylab FX8 Energy Platform (VLFX8GEN) software version 1.1.0 and below.
The good news is that Medtronic has already released security patches for the FT10 platform and the fixes for the FX8 platform are expected to be released in early 2020.
CISA’s advisory provides the following recommendations to minimize the risk of exploitation of these vulnerabilities:
The vendor recommends users to only connect these devices to the hospital network when necessary.
“Medtronic recommends that surgeons and nurses continue to use these devices as intended. Customers should maintain good cyber hygiene practices by only connecting these devices to the hospital network when necessary and shutting them down between uses until the new software update is complete,” reads the advisory published by the vendor.
A separate advisory published by DHS warns of two other vulnerabilities affecting Valleylab FT10 Energy Platform (VLFT10GEN) version 2.1.0 and lower and version 2.0.3 and lower, and Valleylab LS10 Energy Platform (VLLS10GEN) version 1.20.2 and lower.
“Successful exploitation of these vulnerabilities may allow an attacker to connect inauthentic instruments to the affected products by spoofing RFID security mechanisms.” reads the advisory. “This may lead to a loss of
Medtronic already released updates that address both vulnerabilities.
Necessary cookies are absolutely essential for the website to function properly. This category only includes cookies that ensures basic functionalities and security features of the website. These cookies do not store any personal information.
Any cookies that may not be particularly necessary for the website to function and is used specifically to collect user personal data via analytics, ads, other embedded contents are termed as non-necessary cookies. It is mandatory to procure user consent prior to running these cookies on your website.